The following Specifications assume a one year calibration cycle and an operating temperature of 64F to 82 F (18C to 28°C) at relative humidty up to 80% unless otherwise noted. AC Current (Average sensing, calibrated to rms of sine wave)
Range Resolution Accuracy (50Hz – 60Hz)
20A 10mA (3% of reading +5 digits)
2004 100mA (2.5% of reading +5 digits)
400A. 1A (3% of reading +5 digits)
(Overload protection: 400A on all ranges)
AC Voltage (Average sensing. calibrated to ms of sine wave)
Range Resolution Accuracy(50Hz – 500Hz)
450V 1V (2% reading +4 digits)
Input impedance: 9MQ
Overload protection: 450V AC/OC on all ranges.
1 Read the following safely information carefully before attempting to operate or service the multimeter.
2 To avoid damages to the instrument do not exceed the maximum limits of the input values show in the technical specifications tables
3 Never measure current while the test leads are inserted into the input jacks.
4 Do not use the multimeter or test leads if they look damaged. Use extreme caution when working around bare conductors or bus bars.
5 Caution when working with volteges above 60VDC or 30VAC RMS. Such voltages pose a shock hazard.